SCSI2SD SCSI2SD is an approach emulating SCSI hardware with software running in an PSoC microcontroller from Cypress. An ARM Cortex M3 controller is provided as part of the PSoC, and furthermore a bunch of programmable system parts, e.g. usable for interfaces etc. Focus is on mass storage, so SCSI2SD can emulate hard drives and other mass storage devices, all on a SD card. Main website is http://www.codesrc.com/mediawiki/index.php/SCSI2SD . Alternative PCB layout, specialized for Powerbook hardware, but can be used for general SCSI purposes too: https://www.
gnuplot is a nice tool for plotting data. Start gnuplot to enter plot commands. I tried this tool when analyzing a strange behaviour of a Zener diode. In the end it turned out that there was no strange behaviour. Data file. Values separated with whitespaces: Current [mA] Vin Vzener 0.100 2.420 2.396 0.200 2.710 2.652 0.300 2.894 2.862 0.400 3.038 2.911 0.500 3.158 2.997 0.600 3.260 3.068 0.700 3.359 3.130 0.
5 1/2 DMM from the 80ies. It has a hidden 6 1/2 digit mode. My meter does not have the True RMS module for AC. But it has a connector that emits the measurements via RS232 (not tested). While all date codes from the digital board are from 1984, I found also some chips from 1988. So the meter is from 1988 or later. It looks like the meter was in “official” use until 2023, because there is a safety sticker from this year.
HP 6112A DC Power Supply is a precision power supply for 0-40 volts and 0-500mA current. It has a 5 digit digital input to select the precise output voltage. The HP611x is a whole series of such PSUs with different output voltages. I bought this one after I was using the HP6111A for some time and was very happy with it. See here for details on that other PSU. This unit has the crowbar option.
This is a medium size UV EPROM Eraser. It can delete up to 25 chips at once in a removable tray 23x10cm. From datasheet, it has ~17.000µW/cm2 UV intensity. Wavelength is 254nm. It has a large replaceable UV grid tube. Erasing time required from the datasheet is between 5.9 and 14.7 minutes, depending on required erasing energy of the chip to erase (usually given in Watt * seconds/cm2). Tray opened, can fit 25 chips or while PCBs: The mechanical 0-60min timer was loose, I had to fix it: A very nice vintage mystery chip I found inside the eraser: I tried to get some knowledge in that chip: https://www.
ITT teddy 4 came as not working together with another radio I bought from an ebay seller. This radio is from early seventies, I estimate. The radio is medium-sized, not a small one, but also not a large one. Sound is quite impressive, it has more bass than I expected. Sound, compared to e.g. ITT Junior automatic, is much better. This picture was done after repair and cleaning all caging parts.
This is a great DMM from the around 1979. It is called “Microvolt Digital Multimeter” because it has a resolution down to 1 microvolts. Even more interesting is the current range, going down to 1nA (nanoampere) resolution. This is due to the DC range 20 mV (20.000mV) and DC current range 20 µA (20.000µA). Finally, it has a 1mOhms resolution on 20.000 ohms range. This multimeter is great to measure small currents, e.
HP 6111A DC Power Supply is a precision power supply for 0-20 volts and 0-1 ampere current. It has a 5 digit digital input to select the precise output voltage. The HP611x is a whole series of such PSUs with different output voltages. I have also a HP6112A, going upo to 40 volts. See here for details on that other PSU. This is a device from the 60ies or 70ties. I could not find any date code for my device.
Trapider TE-15 Grid-Dipper. Frontplatte mit Frequenzeinstellscheibe, Anzeigeinstrument sowie Frequenzeinstellknopf und Ein/Aus- / Empfindlichkeitsregler: Originalverpackung nur in Teilen vorhanden. Man sieht das Gerät, Spulen für die unterschiedlichen Frequenzbereiche sowie Ohrhörer und Anleitung. Zwei der Spulen, hier für 40-140 Mhz und 120-280 Mhz: Das überschaubare Innere: Transistoren 2SB422: Hier eine Germaniumdiode: Ein Scan der Anleitung: Das Gridmeter in Nutzung. Es ist eine Frequenz von 700 Khz eingestellt. Die Meßspule liegt direkt über einer Spule eines Testaufbaus (Meissner Oszillator).
Darstellung der LC Oszillator Grundschaltungen, aufgebaut mit Bausteinen des Scolatron Baukastensystems von Phywe. Betrachtung des verwendeten Schwingkreises Der Schwingkreis ist in Colpitts- und Hartley-Schaltungen gebildet aus einer Spule mit nominal 200 µH mit Mittelanzapfung und zwei Kondensatoren 47nF und 10nF in Serie. Mit LCR-Meter ermittelte Werte: 47nF Kondensator gemessen mit 46.2 nF (bei 1Khz) 10nF Kondensator gemessen mit 10.9 nF (bei 1Khz) Der Gesamtwert des Kondensators im Schwingkreis ist daher 8.